A 13 by 13 inch adaptive wall test section was installed in the 0.3 Meter Transonic Cryogenic Tunnel circuit. This new test section is configured for 2-D airfoil testing. It has four solid walls. The top and bottom walls are flexible and movable whereas the sidewalls are rigid and fixed. The wall adaptation strategy employed requires the test section wall shapes associated with uniform test section Mach number distributions. Calibration tests with the test section empty were conducted with the top and bottom walls linearly diverged to approach a uniform Mach number distribution. Pressure distributions were measured in the contraction cone, the test section, and the high speed diffuser at Mach numbers from 0.20 to 0.95 and Reynolds numbers from 10 to 100 x 10 (exp 6)/per foot.